Can I use a “UPS” Battery backup to protect my X-ray device?
No – Vatech does not recommend using a “UPS” or known as Battery backup on your X-ray machine. These devices are run off battery and typically can not keep up with the power demand; typically when used will cause a long tone or beeping sound during capturing which indicates it is not able to keep up with the voltage/amperage requirements of the device.
Please view the following topic for Vatech Recommended devices.
Notice: The usage of such devices will cause damage to the equiptment and possibly void manufacture warranty.